the pressure that can be applied by the Honan Intraocular Pressure Reducer.
Preparation For Use, Reusable
Clean and disinfect the Headband, Bellows and Tubing, (parts that touch the patient’s
face) using a recognized germicidal solution. Wipe clean. Do not immerse. Do not
autoclave or gas sterilize.
CAUTION: Do not sterilize the Pressure Gauge or Bulb Air Pump. They could be
damaged in the process.
Positioning the Pressure Reducer
Tape upper lid closed to avoid touching cornea. Place a Sterile Eye Pad or 4” x 4” gauze
over the eye and position the soft pneumatic Bellows over the pad and the eye, securing
it loosely in place with the adjustable Headband.
Apply Headband loosely enough to prevent putting pressure on eye before inflating
Bellows, but tightly enough to apply desired pressure when inflated.
Before inflating Bellows, read the Guidelines For Using The Pneumatic Eye Softener.